Høyfrekvens-forsterker HV10 (HF 58B, 59B)
Increases the sensitivity of the HF-analyser by a factor of 10 (= 10 dB), that is nominal 0.001 µW/m²*. Adaptor plug to be connected to the antenna input socket, remote-fed by the respective basic device. It is especially recommended for the HF59B for measurements in the highly sensitive measuring range of the radar and UMTS mode.
– Future-proof: Comprises the full frequency range of the HF59B.
– In order to avoid measurement errors due to intermodulations, a high pass filter matching to the antenna (such as the HP800) should be applied when measuring with the log.-per. antenna.
– *The actual sensitivity is limited by the device-specific residual noise of the HF-analyser.
– Accuracy: Return loss: 0 Hz (DC) < 5 Ohm, 27 MHz – 2600 MHz = 15 dB +/- 1 dB, 2600 MHz – 3000 MHz = 15 dB – 1 dB bis 15 dB – 4 dB
Reflection loss: 27 MHz – 3000 MHz < -6 dB
– Weight: 0,03 kg
Scope of delivery:
– tightening aid